Class: Instrumentation Manufacturer: Kajaani Model No: SD Part Number: Qty: 1. Description: Automatic Sample Valve. follow-up through a communicator or Kajaani interface via Ethernet. Three operating modes: • Manually initiated. • Collection timer 24/7. STEC SEC Mass Flow Controller. ₪. METSO KAJAANI SD TRANSMIITER USED. ₪. MKS Flow Meter (N2) – Model ALS3BV.
|Published (Last):||25 November 2011|
|PDF File Size:||8.40 Mb|
|ePub File Size:||1.88 Mb|
|Price:||Free* [*Free Regsitration Required]|
If the material is not indicated on a component, see sx5002 device manuals for further information. Use two-pair instrumentation cable for the control cabling. Neither receipt nor possession thereof confers or transfers any right to reproduce or disclose the document, any part thereof, any physical article or device, except by written permission from the manufacturer. Tighten the hex socket screw properly.
Learn Sd0502 – opens in a new window or tab. Ripley, Tennessee, United States. Also install a pressure regulator and manometer Open the top cover of the sampler and make sure that the piston shaft seal does not leak. Replacing the seals of sample delivery valve See drawing E V1.
Air inlet to sampling device Choosing the sampling device type. Replacing the bearing bush and O-ring This operation requires a process shut-down — the sampling device must be removed from the process.
KURZ 454FTB-16-HHT Single Point Insertion Thermal Mass Flow Transmitter
Materials posing environmental hazards These materials must be disposed of as specified by currently valid local legislation and regulations. The air supply pressure must be Check which measuring channel you will be working with, and set it OFF at the analyzer.
Learn More – opens in a new window or tab International shipping and import charges paid to Pitney Bowes Inc. Learn more – opens in a new window or tab. Removing the bearing bush and O-ring. Maintenance every 2 weeks SD process coupling H V1. This seller is currently away until Jan 02, Selecting the correct sampling device 6.
Remove the old O-ring part 16 from the bottom of the bearing bush. Turn the tool by its arms and screw it into the bearing bush as shown in Fig.
There are 1 items available. Piston rotation guard Condition see all Condition. Piston rotation guard, kjaani to the left d Hose coupling 1 for 25 mm hose Image not available Photos not available for this variation. Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab Add to watch list.
Pull the piston out. Checking the actuator diaphragm. SD installation, operating, service – 5. Connect the instrument air hose. See each listing for sd50002 shipping options and costs.
מדי זרימה נוזלי והמונית: פשוט לקנות באיביי בעברית – זיפי
Set the measuring channel ON at the analyzer. Position kajaani flow opening. Requires a process shutdown — sampling device must be removed from the coupling for this replacement! The flow opening of the piston opens more when the piston moves a longer distance. Mounting collar Support pipe, e. Connection box Washing unit 2 Opening in measuring unit bottom for cable inlet Fig.
Kajaanj the rotation guard 8 along the piston shaft 2 as far left as it goes. Installing the closing piston: